Tealth External Irrigation Low Speed Contra Angle Dental Handpiece
|FOB Price:||US $16 / PCS|
|Min. Order:||1,000 PCS|
|Min. Order||FOB Price|
|1,000 PCS||US $16/ PCS|
|Transport Package:||Plastic Box|
|Payment Terms:||T/T, Western Union, Paypal|
- Model NO.: 1024CH
- Certification: TUV, ISO, SGS, CE
- Light Way: Without Light
- Usage Times: Non-Disposable
- Loading Way: Wrench Type
- Exhaust Way: Exhaust Outside
- Connection: Without Quick Connector
- Noise: Less Than 70dB
- Specification: Aluminum, stainless steel
- HS Code: 90184100
- Condition: New
- Material: Aluminum
- Water Way: Outer Channel
- Application: Adult, Child
- Head Diameter: Middle Head
- Number of Holes: Two Holes
- Rotating Speed: Low-Speed Handpiece
- Trademark: TEALTH
- Origin: China
Name: smooth external 1:1 contra angle
push button , easily take it out , No fly out.
Diameter meter is φ2.35mm
External water spray system.
smooth outlooking , conform to ergonomic design.
After each treatment,Sterilizing handpiece in autoclave according to the following steps:
1.Remove the bur and brush clean handpiece, clean outside of handpiece with an alcohol solution.
2.Lubricate handpiece, each times drop two drops inside of air inlet pipe.
3.Place into autoclaving pouch and mark.
4.Autoclave the handpiece for 12 min. at 135ºC.
It shall be sterilized before handpiece is used at the first time;sterilization temperature is 135ºC and last for 12 minutes.
After using,Please follow the instruction to sterilize the handpiece:
1 extract the bur from the handpiece,use the soft brush to clean the dirt and wash with disinfection of alcohol cotton.
2 lubricate the handpiece,pour two drops into air tube one time.
3 wrap it with sterilization package and mark.
Please do not use the acid solution to scrub, inject or soak the handpiece.
Please do not autoclave the handpiece with residual chemical substances.
Handpiece should be packed well and place at clean and dry room without dust, sulfur and salt.
Please do not touch the handpiece immediately after autoclaving.