Tealth Deatchable 20: 1reduction Surgery Implant Contra-Angle Handpiece
|FOB Price:||US $80 / PCS|
|Min. Order:||1,000 PCS|
|Min. Order||FOB Price|
|1,000 PCS||US $80/ PCS|
|Payment Terms:||T/T, Western Union, Paypal|
- Model NO.: 3201CH
- Certification: TUV, ISO, SGS, CE
- Light Way: Without Light
- Usage Times: Non-Disposable
- Loading Way: Gland Type
- Exhaust Way: Exhaust Outside
- Connection: Without Quick Connector
- Gear Ratio: 20:1 Reduction
- Trademark: TEALTH
- Specification: Aluminum, stainless steel
- HS Code: 90184100
- Condition: New
- Material: Stainless Steel
- Water Way: Outer Channel
- Application: Adult
- Head Diameter: Middle Head
- Number of Holes: Two Holes
- Rotating Speed: Low-Speed Handpiece
- Noise: Less Than 70dB
- Transport Package: Plastic Box
- Origin: China
TELATH 3201CH detachable 20:1 implant handpiece
Torque force:Up to 55 N.Cm
Rotation speed :5~2000RPM
Detacheable head design, easily to completely clean up head.
Ergonomics coating ,more hygeian.
Sterilization :Up to 135 ºC
Applicable bur :Φ2.345-2.355mm
Fitting with E type motor , can be avaible to match up with Swiss, Japan, Korea famous brand implant system.
After each treatment,Sterilizing handpiece in autoclave according to the following steps:
1.Remove the bur and brush clean handpiece, clean outside of handpiece with an alcohol solution.
2.Lubricate handpiece, each times drop two drops inside of air inlet pipe.
3.Place into autoclaving pouch and mark.
4.Autoclave the handpiece for 12 min.At 135ºC.
Please do not use the acid solution to scrub, inject or soak the handpiece.
Please do not autoclave the handpiece with residual chemical substances.
Handpiece should be packed well and place at clean and dry room without dust, sulfur and salt.
Please do not touch the handpiece immediately after autoclaving.